## Wednesday, October 31, 2018

### Dissimilar Metal Junctions (How Thermocouples Work)

When two dissimilar metal wires are joined together at one end, a voltage is produced at the other end that is approximately proportional to temperature. That is to say, the junction of two different metals behaves like a temperature-sensitive battery. This form of electrical temperature sensor is called a thermocouple:

This phenomenon provides us with a simple way to electrically infer temperature: simply measure the voltage produced by the junction, and you can tell the temperature of that junction. And it would be that simple, if it were not for an unavoidable consequence of electric circuits: when we connect any kind of electrical instrument to the thermocouple wires, we inevitably produce another junction of dissimilar metals. The following schematic shows this fact, where the iron-copper junction J1 is necessarily complemented by a second iron-copper junction J2 of opposing polarity:

Junction J1 is a junction of iron and copper – two dissimilar metals – which will generate a voltage related to temperature. Note that junction J2, which is necessary for the simple fact that we must somehow connect our copper-wired voltmeter to the iron wire, is also a dissimilar-metal junction which will also generate a voltage related to temperature. Further note how the polarity of junction J2 stands opposed to the polarity of junction J1 (iron = positive ; copper = negative). A third junction (J3) also exists between wires, but it is of no consequence because it is a junction of two identical metals which does not generate a temperature-dependent voltage at all.

The presence of this second voltage-generating junction (J2) helps explain why the voltmeter registers 0 volts when the entire system is at room temperature: any voltage generated by the iron-copper junctions will be equal in magnitude and opposite in polarity, resulting in a net (series-total) voltage of zero. Only when the two junctions J1 and J2 are at different temperatures will the voltmeter register any voltage at all.

Reprinted from "Lessons In Industrial Instrumentation" by Tony R. Kuphaldt – under the terms and conditions of the Creative Commons Attribution 4.0 International Public License.

## Thursday, October 25, 2018

### Hotfoil-EHS Heat Treating Power Consoles

Precise control over your pre-weld and post-weld heat treatment parameters are critical. Accurate temperature control, specific soak times, uniformity, and controlled heat up and cool down times are required to ensure strong welds. Hotfoil-EHS power consoles are designed to provide the best control, easiest user interface, and longest lasting operation, even in the toughest environments. Using only the highest quality components, Hotfoil-EHS power consoles are field-tested and application proven. Control systems can be specified with or without recorders or ramping controllers, and are standardly available in 6, 9, 12, 18, and 24 zone configurations.

https://hotfoilehs.com
609-588-0900

## Thursday, October 11, 2018

### The ICE Advanced Heat Treatment Control System

The ICE IS System, is developed for precise, reliable and efficient heat treatment control. It consists of IS controllers and ISPort software. With ISPort software you can define process parameters as temperatures, rates, tolerances etc; operate and control one or many processes from one or several controllers; edit PID values; fill in needed information, ex. customer info, work info etc.; print all work documents and heat treatment certificates. For more information contact Hotfoil-EHS,

https://hotfoilehs.com/icestar
609-588-0900